Location of L1104
and 11108, R1150
e. Switch the band selector to band C. Turn the INT REF
FREQ control on.
f. Push the LOCK CHECK button. Check for beat signal
displays as the FINE RF CENTER FREQ control is rotated
through its range.
27. Adjust 1 MHz REF FREQ Range
a. Equipment set up is given in step 26.
b. Apply the output from the 1 MHz MARKERS OUT
signal through a 10 attenuator to J120 on the honeycomb
assembly as follows: Connect a GR to Sealectro adapter
(such as a P6040 probe cable), a GR to BNC adapter and a
20 dB attenuator, (see Fig. 6-32) between the 1 MHz MARK-
ERS OUT connector and J120. Disconnect the Sealectro
connector from J120.
c. Set the DISPERSION to 100 kHz/Div and position a
1 MHz marker to the screen center with the IF CENTER
d. Adjust L1108 (Fig. 6-34) for a positive oscillator start
when the INT REF FREQ control is turned from its OFF to
ON position. The signal position on the screen should be
consistent as the INT REF FREQ control is switched from
just below the state of free running avalanche. Free running
OFF to the initial ON position. (The control must not be
avalanche, appears as an RF burst signal between the 1 MHz
pulses as shown in Fig. 6-33C; or with the INT REF FREQ
will be changed.]
control in the OFF position, the free running avalanche
transistor signal will feed through to the 1 MHz MARKERS
e. Rotate the INT REF FREQ control through its range
Check the total frequency shift of the internal reference
2 MHz burst. If free running avalanche does not occur, turn
1 MHz marker.
the adjustment fully clockwise.